Stainless Steel Autoclave Sterilizer
This product has the advantages of novel and beautiful shape, reasonable structure, complete functions, rapid heating and thorough sterilization. It is suitable for medical, scientific research, food, and other units to sterilize surgical instruments, dressings, glassware, rubber products, food, liquid medicine, and culture media.
1. Steam heat penetration is strong.
2. High sterilization efficiency.
3. Sterilization time is short.
4. The sterilization process does not produce any chemical or physical contamination.
5. Sterilization equipment has few control parameters and stable operation.
1.Type B is a common type: it can be used for gas heating or electric heating.
2.Type A is a B-type improved type: the power switch and the water discharge nozzle are added to facilitate the control of heating and drainage speed.
3.Type D is intelligent temperature control type: with automatic constant temperature, constant temperature timing, low water level protection and so on.
Rated Working Pressure
AC110V 60Hz/AC220V 50Hz
Packing & Delivery
Packing:Carton,Plywood Package,Neutral Packing With Export Standard Or Customized According to Customer's Requirement.
Delivery:It depend on the quantity ordered.Normally within 15-20 business days after receiving the payment.
Common Malfunctions Part 1
1. When the instrument is used, the heating function fails and the steam cannot be generated normally.
Fault: When the instrument is running, the heat indicator on the panel is on, but the temperature indication does not rise and the room temperature is maintained.
Treatment method: Open the side cover of the instrument, the panel display is normal, indicating that the 220V power supply normally enters the control circuit. Remove the wire connected to the heating pipe and check the resistance of the heating pipe with a multimeter. Generally, the resistance is between 8 and 15 Ω. If the resistance is too large, the internal heating circuit is broken and replaced in time. Since the heating pipe is burnt out for a reason, the solid state relay that controls the heating pipe should be inspected regardless of whether the heating pipe is good or bad in the first step. In the case of power-on, the relay input voltage should be around 12VDC, and the output voltage should be around 220VAC. If the input is normal and the output is abnormal, the relay must be replaced. If the input is abnormal, check the load output control part of the control circuit.