Medical Autoclave Sterilizer
This product is a device that sterilizes articles, instruments, liquids, etc., using pressurized saturated steam. It is suitable for use in laboratories in the medical, food and other industries. It uses high-pressure hot-saturated steam to sterilize articles.
1.All stainless steel deep drawn seamless construction.
2.Light, compact, portable, easy to use top loading autoclaves.
3.Equipped with two stages over pressure protection system incorporating a
4.calibrated continuous bleeding pressure stopcock and pressure release weight valve.
5.Extended life gasket & heating element.
6.Equipped with heat resistant handles and steam release / control valve.
7.For assurance of quality each autoclave is tested to the parameter of successful sterilization.
1.Type B is a common type: it can be used for gas heating or electric heating.
2.Type A is a B-type improved type: the power switch and the water discharge nozzle are added to facilitate the control of heating and drainage speed.
3.Type D is intelligent temperature control type: with automatic constant temperature, constant temperature timing, low water level protection and so on.
Rated Working Pressure
AC110V 60Hz/AC220V 50Hz
Packing & Delivery
Packing:Carton,Plywood Package,Neutral Packing With Export Standard Or Customized According to Customer's Requirement.
Delivery:It depend on the quantity ordered.Normally within 15-20 business days after receiving the payment.
We offer inquiry and consulting support,sample testing support, training how to use the device,engineers provide technology support.Our warranty period is one year.
1. When placing sterilized articles, it is strictly forbidden to block the safety valve and the venting valve. The vacancies must be left to ensure that the air is unblocked, otherwise the container may burst.
2. Do not exceed 3/4 of the container when placing the liquid.
3. Do not sterilize the items with different sterilization standards.
4. After the sterilization is finished, if the pressure gauge indicator needle has returned to zero position but the lid is not easy to open, the deflation valve tap can be placed in the deflation position, so that the outside air enters the pot, and the vacuum can be removed before the vacuum is removed. cover.
5. The pressure gauge should be repaired in time as the pressure indicator is incorrect or cannot return to zero.
6.Always keep the equipment clean and dry, can extend the life of the equipment. Rubber seals are prone to aging and should be replaced regularly.
7. The reliability of the safety valve should be checked regularly. When the working pressure exceeds 0.165MPa, it does not jump. A new safety valve needs to be replaced.