Portable Table Top Sterilizer
The principle of using microorganisms to kill microorganisms by damp heat is that steam has a strong bactericidal action, which can increase the water content of the bacteria protein, making it easy to coagulate due to heat, and accelerate the death process of microorganisms. It is suitable for sterilization of medical, scientific research, food, and other unit items.
The new generation of portable autoclaves developed by our company has successfully broken through the cumbersome and backward operating procedures of the old portable sterilizers and the unsafe closed-cover structure.
The new sterilizer is advanced in performance, safe and reliable. It has a quick and easy sealing method and automatic discharge of cold air. Safety locks, automatic lock cover and mis-operating cover are not added to the safety control.
Made of high-quality stainless steel, the sterilization index has reached the national medical operation level standard. It is used by various medical and health institutions, colleges, food chemicals, biological research institutes for equipment, materials, food utensils, liquid medicine, culture media and other items. Ideal for sterilization. With water cut protection, pressure self-locking device, safety interlocking device, self-expanding seal.
1.Type B is a common type: it can be used for gas heating or electric heating.
2.Type A is a B-type improved type: the power switch and the water discharge nozzle are added to facilitate the control of heating and drainage speed.
3.Type D is intelligent temperature control type: with automatic constant temperature, constant temperature timing, low water level protection and so on.
Rated Working Pressure
AC110V 60Hz/AC220V 50Hz
Packing & Delivery
Packing:Carton,Plywood Package,Neutral Packing With Export Standard Or Customized According to Customer's Requirement.
Delivery:It depend on the quantity ordered.Normally within 15-20 business days after receiving the payment.
Common Malfunctions Part 2
1. The instrument leaks and the pressure cannot meet the requirements.
Fault: The instrument leakage is common in the sealing rubber ring of the instrument cover, which is caused by the sealing rubber ring aging or the rotation of the upper cover.
Treatment method: Take out the sealing rubber ring and reinstall it. If the damage is not serious, it can be treated with talcum powder. If it is aging or damaged seriously, it needs to be replaced in time.
2. The safety valve works abnormally, and the pressure is too high to relieve pressure.
Fault: The safety valve is the last protective barrier of the autoclave. It automatically protects the safety of the instrument and the operator when the instrument is working abnormally and continuously heating so that the pressure is greater than the safe value.
Treatment method: Manually open the safety valve pressure relief when the pressure is lower than the safety value to prevent the safety valve from rusting for a long time. If the damage is serious, it must be replaced and re-commissioned by a professional.
3. The instrument can not be properly vented and drained, resulting in a large difference between pressure and temperature, which cannot be properly sterilized.
Fault: Generally, the bottom of the automatic self-control autoclave has exhaust and discharge three-way valves. Failure to properly exhaust will result in high pressure indication of the instrument and easy danger.
Treatment method: Open the side cover of the instrument, remove the drain line and the three-way valve to clear. Corrosion must be replaced severely, and the exhaust pipe and drain valve should be flushed regularly during the use. In order to prevent this problem from reappearing, the distilled water should be added to the sterilizer and replaced regularly. Once, add tap water and replace it once a day).